IEEE Standards Interpretation for IEEE Std 743™-1995 IEEE Standard Equipment Requirements and Measurement Techniques for Analog Transmission Parameters for Telecommunications
Copyright © 2001 by the Institute of Electrical and Electronics Engineers, Inc. 3 Park Avenue New York, New York 10016-5997 USA All Rights Reserved.
Interpretations are issued to explain and clarify the intent of a standard and do not constitute an alteration to the original standard. In addition, interpretations are not intended to supply consulting information. Permission is hereby granted to download and print one copy of this document. Individuals seeking permission to reproduce and/or distribute this document in its entirety or portions of this document must contact the IEEE Standards Department for the appropriate license. Use of the information contained in this document is at your own risk.
IEEE Standards Department Copyrights and Permissions 445 Hoes Lane, Piscataway, New Jersey 08855-1331, USA
20 August 2001
Interpretation Request #1
Topic: Clerical error in the tabulation of loss vs. frequency for the D Filter Relevant Clause: 9.1 D Filter Relevant Figure: Figure 17 Relevant Table: Table 30
In Figure 17 on page 83 of IEEE Std 743-1995, a tabulation of loss vs. frequency for the D Filter is given. At the bottom of page 85, a chart (Table 30) of pole/zero values for the same filter is given. The problem is that, in the 300-700 Hz Region, the loss calculated using the pole/zero data does not comply with the loss given in Table 30 in the third row in the column labeled "Z-plane poles."
There is an apparent clerical error in Table 30 of IEEE Std 743-1995. The source document specifying the digital poles and zeros, "Contribution IEEE P743/94-47," shows the components on the third lline of the "z-planes poles" column, to be 0.7258+/-j0.1629, and thus the real component number in the IEEE Std 743-1995 (0.7528) is apparently in error.